Application Programming Interface API To help accelerate software development, all instrument drivers follow a consistent programming flow of: For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node. The default value is 1. This error code doesnot have any explanation attached to it either. This may take some trial and error, maybe even some reading on your part to get this to work. The logic analyzer setup runs on a firmware developed by agilent itself. Open these examples using the following procedure.
|Date Added:||27 February 2013|
|File Size:||20.28 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node. Also, I’ve heard of tools that allow you to “draw” the waveform and save it in an image file and then feed that image file into the function generator.
Take note that when the instrument is turned on, the GPIB Address that is assigned to the device is displayed. If everything is correct the power supply will now be changed to 5. Do the same for the burst 33250 and burst period terminals on the left side and bottom of the Configure Burst node.
Then click the green Run arrow at the top left of the VI. After above, I opened “ag33xxx – Getting Started 2. Then in the next step i need to load a file into Pattern Gen module which is inbulit into one of the slot of the LAPG setup. These are the Controls Palette and the Function Palette.
This is how the location of certain items will be addressed. Wire the purple instrument wire and yellow error wire through the Configure Burst node.
If I changed Freq, labvew changed of waveform.
labview • View topic • How to use an binary stream file to generate Arb from Agilent A?
This error code doesnot have any explanation attached to it either. The first thing to learn about LabVIEW programming is the type of interface that the devices are connected. Any information that appears in parenthesis after the name of the terminal is the default setting.
The labvied developed by National Instruments labvieww widely used in industry for a variety of applications. All the new indicators will be labveiw on the Diagram and the Front Panel. Would you please send the code to this forum – then we can discuss about it.
Now make the offset of the Waveform non-zero and observe what happens. Notice that as the mouse is moved over each of the different blocks of the windows, text at the top of the window shows what the block contains.
Keysight 33250A signal generator, 80 MHz
Open the Learning tab by selecting the Learn to Program tile in the Welcome screen or selecting the Learning tab from the top right. I have written the code but its showing Error This takes the error information out of the GPIB Write, if any exists and places it into separate categories. The next step is optional and is not required, but could help in laview errors that occur in the communication with the device. The default value is 1.
For instance, later this term you will be asked to use LabVIEW to make voltage measurements for a circuit.
Automate Signal Generation From a Keysight/Agilent 33000 Function Generator With LabVIEW NXG
Converting Labview Test Executive 5. The Front Panel and the Diagram are shown in Figures 1 and 2 respectively. Right click terminals and choose Change to control to create a control wired to that terminal. NI develops and supports thousands of instrument drivers for third party instruments to help labviwe and scientists automate measurements. I used AgilentD scope to check AgilentA output directly via coaxial-cable. You can find these instrument drivers in the Instrument Driver Network.
You can configure the Configure Burst node with several inputs. A Sequence Box performs all labbview contained in a frame before moving on to the next frame.
The first is the Front Panel and this is where all the controls and indicators are located.